Popular products by category
Rigaku nano3DX is a true X-ray microscope (XRM) with ultra-wide field of view, 25X larger volume than comparable systems, and three X-ray wavelengths for different matrices.Read more...
HyPix-3000 is a next-generation two-dimensional hybrid pixel array semiconductor detector designed specifically to meet the needs of the home lab diffractionist.Read more...
Analytical solutions by industry
New papers of interest
Special Feature: Pharmaceutical Analysis (5): Analysis of trace impurities in pharmaceutical products using polarized EDXRF spectrometer NEX CG.Read more...
Thickness, density and roughness for multilayers on wafers
Used to determine thickness, density and roughness for single and multilayer stacks on semiconductor wafers, XRR analysis can be performed on both crystalline and amorphous materials. When X-rays are applied to a material's flat surface at grazing angles of incidence, total reflection will occur at or below a certain angle, θc. This angle is exceedingly small and is referred to as the critical angle. The angle varies depending upon the electronic density of the material. The higher the incident X-ray angle relative to the critical angle, the deeper the X-rays transmit into the material. With a material whose surface is ideally flat, the reflectivity suddenly decreases at angles above the critical angle in proportion to θ-4.
If the material surface is rough, it causes a more drastic decrease in reflectivity. If such a material, serving as a substrate, is evenly overlaid with another material having a different electronic density, then reflected X-rays from the interface between the substrate and the thin film as well as from the free surface of the thin film will either constructively or destructively interfere with each other—resulting in an interference induced oscillation pattern. To a first approximation, the intensity scattered by a sample is proportional to the square of modulus of the Fourier transform of the electron density. Thus the electron density profile can be deduced from the measured intensity pattern, and subsequently the vertical properties (layer thicknesses) as well as the lateral properties (roughness and correlation properties of interfaces or lateral layer structure) characterizing multilayers can be determined. Specifically, film thickness can be determined from the periodicity of the oscillation and information on the surface and interface from the angular dependency of the oscillation pattern's amplitude.
Advanced state-of-the-art high-resolution XRD system powered by Guidance expert system software
High-performance, multi-purpose XRD system for applications ranging from R&D to quality control
World's most powerful θ/θ high-resolution X-ray diffractometer features an in-plane diffraction arm
Process XRR, XRF, and XRD metrology tool for patterned wafers; up to 300 mm wafers