Popular products by category
Rigaku nano3DX is a true X-ray microscope (XRM) with ultra-wide field of view, 25X larger volume than comparable systems, and three X-ray wavelengths for different matrices.Read more...
HyPix-3000 is a next-generation two-dimensional hybrid pixel array semiconductor detector designed specifically to meet the needs of the home lab diffractionist.Read more...
Analytical solutions by industry
New papers of interest
Special Feature: Pharmaceutical Analysis (5): Analysis of trace impurities in pharmaceutical products using polarized EDXRF spectrometer NEX CG.Read more...
Determination of Metals in Copper Concentrate by Advanced Correction Method for Fused Beads
This application note will introduce an advanced correction technique for fusion method and application for copper concentrate analysis.
Global copper supply and consumption has been growing in the last few decades. Copper is one of the important base metals for development of modern nations and then it plays a significant role as an essential industrial material for construction, electronics, machinery, transportation and so on.
Copper ore and copper concentrate, which are mainly sulfide, contain variety of minor elements, which are valuable elements as byproducts, harmful to quality in copper metal products and hazardous to environment and human health. Therefore, rapid and accurate determination of various metals of major and trace amount in copper ore and concentrate is important for process control in the beneficiation and smelting.
X-ray fluorescence spectrometry is the best analysis method to monitor ore grade and impurity with high precision, accuracy and rapidity.
The fusion method is an effective sample preparation technique for accurate analysis of a wide variety of powder samples including minerals and ores by XRF spectrometry since it eliminates heterogeneity due to grain size effects and mineralogical composition. In order to take advantage of the fusion method, however, appropriate corrections for errors arisen from fusion processes are required. Possible errors in the fusion process are weighing inaccuracies, loss on ignition (LOI), gain on ignition (GOI), decomposition of oxidizing agent and evaporation of flux.
Rigaku ZSX Primus III+ delivers rapid quantitative determination of major and minor atomic elements, from oxygen (O) through uranium (U), in a wide variety of sample types — with minimal standards. Read more...