Powder X-ray diffractometers can readily measure bulk samples provided techniques are devised for sample preparation. In the case of samples in sheet form, such as paper or film, a diffraction pattern with no angular error can be obtained, as shown in Fig. 1, by: (1) Cut the sample to the same size as the aluminum sample plate, (2) Stacking the sample and sample plate, and (3) Set the sample plate into the sample clip. If the sample has enough rigidity and thickness, it should be placed directly onto the sample clip, without using the aluminum sample plate.
Fig. 2 shows the X-ray diffraction pattern for two types of paper (magazine, white envelope). It is evident that both samples contain cellulose (the main component of paper), but each has different materials, such as clay minerals (e.g., kaolinite), calcium carbonate (calcite, aragonite) and titanium dioxide (rutile, anatase), which are used for surface treatment.
Fig. 2: X-ray diffraction pattern of paper
Apparatus conditions: MiniFlex600 (F.F tube 40 kV, 15mA), Detector: D/teX Ultra, Slit conditions: DS = 1.25°, SS = 8 mm, RS = 13 mm, Incident side and receiving side Soller slit: 5°, Incident height limiting slit = 10 mm
Scan range: 2θ = 5 ~ 55°, Step width: 0.02°, Scan speed: 20° / min. (about 3 min.)
New 6th-generation general purpose benchtop XRD system for phase i.d and phase quantification
Advanced state-of-the-art high-resolution XRD system powered by Guidance expert system software
Highly versatile multipurpose X-ray diffractometer with built-in intelligent guidance
High-performance, multi-purpose XRD system for applications ranging from R&D to quality control
Laboratory micro-spot XRD residual stress analysis with both iso- and side-inclination methods
2D X-ray detector with latest semiconductor technology designed for home lab diffractometers