Generally, characterization of texture in metal materials using X-ray diffraction is carried out using pole figures of multiple indices. The Shultz reflection method using the 0-Dimensional (0D) mode is widely used to obtain pole figures. However, the 0D mode requires measurement of pole figures of reflections one-by-one and the collection of background intensity for each alpha position. These results in long measurement times. Intensities at different alpha positions can be collected simultaneously from the intensity distribution on a Debye ring by using a 2-Dimensional (2D) detector. “2D pole figure measurement” allows you to obtain pole figures in much shorter timeframes compared to 0D pole figure measurement, thus accelerating your research.
Advanced state-of-the-art high-resolution XRD system powered by Guidance expert system software
Highly versatile multipurpose X-ray diffractometer with built-in intelligent guidance
New 6th-generation general purpose benchtop XRD system for phase i.d and phase quantification
High-performance, multi-purpose XRD system for applications ranging from R&D to quality control
Windows®-based software suite for Rigaku's X-ray diffractometers
Laboratory micro-spot XRD residual stress analysis with both iso- and side-inclination methods
2D X-ray detector with latest semiconductor technology designed for home lab diffractometers